Test Range 0 - 1mm (0.05in) up to 0 - 20000 mm (800 in.) at steel
Velocity Variable in 1,2,5 sequence or continuously in 1mm(0.05in) increments. Also from 1 to 5000(?s). Velocity 256 to 16000m/s continuously variable.
Probe Zero 0 to 999.999 ?s, continuously variable.
Delay Calibrated delay from 0 -20000mm in 0.05 mm steps at steel velocity (0-400in. in 0.002 in. steps).
Gain 0 to 110dB. Adjustable in 0.1, 0.5, 1, 2, 6, 10, 14 and 20dB steps. Direct access to gain control at all times.
Test Modes Pulse echo and transmit/receive.
Pulser 100V - 350V (450V MS380) square wave pulser.
ActiveEdge TM Unique active pulse control for enhanced near surface resolution and signal response. Replaces traditional damping control.
P.R.F Selectable 5 to 5000 Hz.
Thickness Logging Storage for 8000 thickness readings configured either by Block/Location/Number mode or pre-programmable work sheets in sequential mode. Readings can be exported to MS Excel using optional SDMS software.
Language Support Supports multiple languages. User selectable between English, German, Spanish, French, Dutch, Italian, Russian, Polish, Czech, Finnish & Hungarian. Others available on request.
Mode 1 Signal Monitor
Mode 2 Depth and amplitude of first signal in gate.
Mode 3 Echo-to-Echo distance measurement. (single gate)
Mode 4 Trigonometric display of beam path, surface distance and depth of indication, curve surface correction and X-OFFSET for probe index. Half skip indication on screen.
Mode 5 Gate to Gate distance measurement. (independent gates).
Mode 6 T-Min mode for holding minimum thickness reading. Resolution to 0.01mm (0.001in) for distance measurement or 1% FSH for amplitude measurement. Large display of measurement at top of A-Scan display. Measurement mode selectable between peak and flank.
AGC Automatic Gain Control automatically sets the signal to a level between 10-90% FSH with tolerance between 5-20% accuracy.
DAC DAC defined by up to 10 points and digitally drawn on screen. DAC curves meet requirements of EN 1714, JIS and ASME standards, select able between -2, -6, -10, -12 and -14dB. Amplitude read out selectable between % DAC, % FSH or relative dB.
TCG Time Corrected Gain, also known as Swept Gain. 40dB dynamic range greater than 30dB per microsecond and up to 10 points may be used, setting all signals initially to 80% FSH.
Update Rate 60Hz (NTSC Mode); 50Hz (PAL Mode).
Rectification Full wave, positive or negative half wave and unrectified rf.
Frequency Range 6 narrow bands centred at 0.5 MHz, 1MHz, 2.25MHz, 5MHz, 10MHz and 15MHz. Broad band at 2 MHz to 22MHz (-6dB) and 1MHz to 35 MHz (-20dB).